The skeptical cardiologist picked up an Apple Watch 4 at the Galleria Apple Store in St. Louis today. The Apple employee who retrieved it told me that ECG recording capabilities were expected in the fall. Of course fall began today and it is not at all clear when, if ever, Apple will provide the software update to its AW4 that will provide ECG capabilities.
Fortunately, consumers already have the capability of recording a medical grade single lead ECG with any Apple Watch 2 or 3-using the KardiaBand from AliveCor.
Apple has hubristically proclaimed the AW4 as the ultimate guardian of our health and while setting it up I was asked if I wanted the watch to notify me if my heart rate dropped below 40 bpm for 10 minutes. Sure! Let’s see how irritating this feature will be.
After setting up the new watch I immediately attached my KardiaBand and installed the Kardia Apple Watch app.
I was able to open the Kardia app and it performed its normal SmartRhythm monitoring but when I tried to record an ECG, alas, nothing happened.
It appears that the KardiaBand does not work with the new Apple Watch 4. Yet.
I was informed by Ira Bahr at AliveCor that their “testing on AW4 is not yet complete. So at present, the device is not supported.”
Now I face a difficult decision-Do I wear my new AW4 with a non KardiaBand wrist band (and no ECG capability) or wear my old Apple Watch with the KardiaBand (and outstanding ECG capability.)
As I pointed out Friday, Apple’s claim that the ECG sensor on their new Apple Watch 4 (available “later this year”) is “the First ECG product offered over the counter directly to consumers” is totally bogus.
AliveCor’s Kardia mobile ECG device was approved by the FDA for over the counter direct to consumer sales on February 10, 2014. Apple had to have known this as they worked with AliveCor to bring the first Apple Watch based ECG device to FDA approval in 2017.
I tried but failed to get AliveCor founder Dr. David Albert’s thoughts on Apple’s disinformation but Yahoo finance was able to speak to Vic Gundotra, the CEO of AliveCor:
Over at the headquarters of AliveCor, a startup based in Google’s hometown of Mountain View, they, too, were surprised by the announcement, CEO Vic Gundotra said in a phone interview on Thursday. Gundotra is a former Googler, widely known as the executive behind the Google+ social network.
Gundotra was also surprised by Apple’s claims of ECG primacy
“We were watching [the announcement], and we were surprised,” Gundotra said. “It was amazing, it was like us being on stage, with the thing we’ve been doing for 7 years,” referring to AliveCor’s product for detecting atrial fibrillation (AFib), a tough-to-spot heart disorder that manifests as an irregular, often quick heart rate that can cause poor circulation.
“Although when they said they were first to go over-the-counter, we were surprised,” he continued. “Apple doesn’t like to admit they copy anyone, even in the smallest things. Their own version of alternative facts.”
One man’s alternative fact is another (less polite) man’s lie.
Gundotra apparently views Apple’s entry as a good thing
“We love that Apple is validating AFib; just wait until you see what AliveCor is going to do next,” he said. “We were a great restaurant in a remote section of town, and someone just opened a giant restaurant right next to us, bringing a lot more attention.”
And as I pointed out previously, the AliveCor mobile ECG device (not the Kardia Band) is significantly cheaper than an Apple Watch and has multiple studies showing its accuracy. Interestingly, Gundotra indicates AliveCor sales has increased after the Apple announcement,.
“Ours is $99, theirs is $399, our sales popped yesterday, big time,”
In March of 2017 AliveCor introduced Kardia Pro, a cloud-based software platform that allows physicians to monitor patients who use the Kardia mobile ECG device.
I have been utilizing the Kardia mobile ECG device since 2013 with many of my atrial fibrillation (AF) patients and have found it be very useful as a personal intermittent long term cardiac monitor. (see here and here)
I signed up for the Kardia Pro service about 3 months ago and all of my patients who purchased Kardia devices prior to March of 2017 have been migrated automatically to Kardia Pro by AliveCor.
Now (post March 2017), patients who acquire a Kardia device must sign up for the Kardia Pro service at $15 per month to connect with a physician.
I think this is money well spent and I’ll demonstrate how the service works with a few examples.
Monitoring Patients With Atrial Fibrillation
I saw a 68 year old man with persistent atrial fibrillation that was first diagnosed at the time of pneumonia in late 2017.
He underwent a cardioversion after recovering from the pneumonia but quickly reverted back to AF. His prior cardiologist offered him the option of repeat cardioversion and long term flecainide therapy for maintenance of normal sinus rhythm (NSR) but he declined.
When I saw him for the first time in the office a month ago I listened to his heart and to my surprise, noted a regular rhythm: an AliveCor recording in the office confirmed he was in NSR. The patient had been unaware of when he was in or out of rhythm
We discussed methods for monitoring his rhythm at this point which include a 24 Holter monitor, a 7 to 14 day Long Term Monitor, a Cardiac Event Monitor and a Mobile Cardiac Outpatient Telemetry device. These devices are helpful and although expensive are often covered by insurance. They require wearing electrodes or a patch continuously and the results are not immediately available.
I also offered him the option of monitoring his AF using a Kardia device with the recordings connected to me by Kardia Pro.
He purchased the device on his own for $99, downloaded the app for his smartphone and began making recordings.
I enrolled him in my Kardia Pro account and he received an email invitation with a code that he entered which connected his account with mine, allowing me to view all of his recordings as they were made.
When I log into my Kardia Pro account I can now view a graphic display of the recordings he has made with color coding of whether they were considered normal or abnormal by Kardia.
The patient overview page also displays BP information if the patient is utilizing certain Omron devices which work with Kardia.
The display shows that after our office visit he maintained NSR for 3 days (green dots) and then intermittently had ECG recordings classified as AF (yellow dots) or unclassified (black).
The more he used the device and got feedback on when he was in or out of rhythm the more he was able to recognize symptoms that were caused by AF.
I can click on any of the dots and six second strips of the full recording are displayed. In the example below I clicked on 2/27 which has both an unclassified recording (which is atrial flutter) and an AF recording
Clicking on the ECG strips brings up the full 30 second recording on a page that also allows me to assign my formal interpretation. In the example below I added atrial flutter as the diagnosis, changing it from Kardia’s unclassified (Kardia’s algorithm calls anything it cannot clearly identify as AF that is over 100 BPM as unclassified.)
The ECG can then be archived or exported for entry into an EHR.
The benefits of this patient being connected
to me are obvious: we now have an instantaneous patient-controlled method for knowing what his cardiac rhythm is doing whether he is having symptoms or not.
This knowledge allows me to make more informed treatment decisions.
The Kardia Pro Dashboard
When I log into kardia pro I see this screen.
It contains buttons for searching for a specific patient or adding a new patient. Adding new patients is a quick and simple process requiring input of patient demographics including email and birthdate.
From the opening screen you can click on your triage tab. I have elected to have all non normal patient recorded ECGS go into the triage tab.
Another patient’s Kardia Pro page shows that he records an ECG nearly every day and most of the time Kardia documents NSR in the 60s. Overall, he has made 773 recordings and 677 of them were NSR, 28 unanalyzed (due to brevity) , 13 unclassified and 55 showing AF.
Monitoring Rate Control In Patients With AF and Reversion Post-Cardioversion
Another patient I saw for the first time recently has had long-standing persistent AF. His previous cardiologist performed an electrical cardioversion a year ago but the patient reverted back to AF in 40 hours. Before seeing me he had purchased a Kardia mobile ECG device and was using it to monitor his heart rate.
After he accepted my email invitation to connect via Kardia Pro I was able to see his rhythm and rate daily. The Kardia Pro chart belowshows his daily heart rate while in atrial fibrillation. We utilized this to guide titration of his rate controlling medications. Such precise remote monitoring of heart rate in AF (which is often difficult to accurately assess by standard heart rate devices) obviates the need for office visits for 12 lead ECGs or periodic Holter monitors.
I performed a second cardioversion on him after which he made daily recordings documenting maintenance of NSR. With this system we can determine exactly when AF returns, information which will be very helpful in determining future treatment options.
Kardia Pro Plus Kardia Mobile ECG Creates Personal Intermittent Long Term Rhythm Monitor
There are many potential applications of the Kardia ECG device beyond AF monitoring (assessing palpitations, PVCs, tachycardia, etc.) but they are all enhanced when the device is combined with a good cardiologist connected to the device by Kardia Pro.
I’ve gotten spoiled by the information I get from my AF patients who are on Kardia Pro now. When they call the office with palpitations or a sense of being out of rhythm I can determine within a minute what their rhythm is wherever I am (excluding tropical beaches and mountain tops) or wherever the patient is (for the most part.)
On the other hand patients who are not on Kardia Pro have to come into the office for 12-lead ECGs. When they call I feel like my diagnostic tools are limited. Such patients usually end up getting one of the standard Long Term Monitoring (LTM) Devices. If I am fortunate, after a few days to weeks , the results of the LTM will be faxed to my office.
I am optimistic based on this early experience with Kardia Pro that ultimately this service in conjunction with the Kardia Mobile ECG device (or similar products) will replace many of the more expensive and inconvenient long term monitoring devices that cardiologists currently use.
The KardiaBand for Apple Watch from AliveCor has delivered on its unique promise of a medical grade single lead ECG recording made by placing your thumb on your wristwatch band.
The ECG recordings are equivalent in quality to those made by their previously available KardiaMobile (see my prior post here.) After more experience with the Band I think the ease of recording is superior to KardiaMobile and the ability to discriminate atrial fibrillation from normal sinus rhythm is similar to KardiaMobile.
By combining either a KardiaBand or a KardiaMobile device with Kardia’s SmartRhythm monitoring system for Apple Watch we now have the promise of personal monitoring to detect atrial fibrillation.
What is SmartRhythm?
SmartRhythm is AliveCor’s term for its system for monitoring your heart rate and activity levels in order to identify when your rhythm is abnormal.
The system “takes your heart rate and activity data gathered from the Apple Watch and evaluates it using a deep neural network to predict your heart rate pattern.”
The heart rate is obtained from the Apple Watch PPG sensor every 5 seconds. If it differs from what is predicted SmartRhythm notifies you to record an ECG.
If you’d like to learn more detail about the development of SmartRhythm and how it functions, AliveCor has an excellent informational piece here.
You can choose to have the Kardia SmartRhythm display come up whenever your Apple Watch awakens. It’s got information on your heart rate and activity over the preceding several hours
The AliveCor FAQ on SmartRhythm stresses that a notification does not always mean an abnormal rhythm. Clearly false positives can and will occur. The first day I wore my KardiaBand I had several of these.
Causes for false positives include exercise that Apple Watch couldn’t detect, stress or anxiety-in other words, situations where your heart rate is higher than predicted by how much activity you are doing.
The long term record of your SmartRhythm recordings resides on your iPhone . Here’s my record for the last week
Note that Kardia , in addition to tracking your heart rate, also shows you by the green, yellow and orange dots, the times that ECG recordings were made.
Green dots indicate recordings classified as normal and yellow as “unclassified.” In my case most of the unclassified recordings were due to heart rate >100 BPM associated with exercise.
There is one orange dot indicating that Kardia felt the ECG showed “possible atrial fibrillation.”
This happened when I took my Apple Watch off my wrist and put it on one of my patients who has permanent atrial fibrillation. I had him push on the KardiaBand sensor to make an ECG recording and it was correctly identified as atrial fibrillation.
Thus far I have had no notifications of “possible atrial fibrillation” while I have been wearing my watch thus the false positive rate appears acceptably low.
How Does SmartRhythm Perform During Exercise?
I checked out SmartRhythm’s ability to predict normal and abnormal heart patterns by wearing it during a session on my indoor bike trainer. The device did a good job of tracking both my heart rate and activity during the workout. You can view the most recent data by viewing your Apple Watch screen during the workout as below
Or for more detailed information you can view the complete history on your iPhone as below
The system accurately tracked my heart rate and activity (although AliveCor lists stationary bike as an activity that may result in false positives). During a session of weights after the aerobic workout despite erratic heart rates and arm movements it did not notify me of an abnormality. I also did 100 jumping jacks (which involves wildly flailing my arms) and the heart rate remained within the predicted boundaries.
What is more remarkable is that I was able while cycling at peak activity to make a very good quality ECG recording by taking my right hand off the handle bar and pushing my thumb down on the KardiaBand sensor on my left wrist.
This recording clearly displays p waves and is sinus tachycardia. It’s unclassified by Kardia because the rate is >100 BPM.
Afib Patient Experience
One of my patients last week, a 70 year old woman with paroxysmal atrial fibrillation, had already set up SmartRhythm monitoring on her Apple Watch.
I have this patient like many of my afibbers utilizing KardiaMobile to check an ECG when they think they are in afib.
However, she, like many of my afib patients, is totally unaware when her heart is out of rhythm. Such asymptomatic patients are alerted to the fact that they are in afib by detection of a rapid heart rate (from a heart rate tracking wearable or BP monitor) or an irregular heart beat (from BP monitor or by someone checking the pulse) or by a random recording of an ECG.
She’s started using SmartRhythm in the hopes that it will provide a reliable and early warning of when she goes into atrial fibrillation.
We discussed the possibility of stopping the flecainide she takes to maintain normal rhythm to test the accuracy of the SmartRhythm system for detecting atrial fibrillation in her but decided not to. She’s on an oral anticoagulant and therefore protected from stroke so development of atrial fibrillation will not be dangerous for her.
I eagerly await the first real world, real patient reports of SmartRhythm’s performance in atrial fibrillation detection.
If there are any afibbers out there who have had an episode of atrial fibrillation detected by SmartRhythm please let me know the details.
We need such anecdotes along with controlled trials to determine how useful SmartRhythm will be as a personal wearable system for detection of afib.
N.B. I’ve copied a nice section from AliveCor’s website which describes in detail the difference between measuring heart rate from the PPG sensor that all wearable devices use versus measuring the electrical activity of the heart with an ECG.
To understand how Kardia for Apple Watch works, let’s start by talking about your heart, how the Apple Watch and other wearable devices can measure your heart rate, and how an ECG is different from the information you get from a heart rate sensor alone.
Your heart is a pump. With each beat of your heart, blood is pumped through your arteries and causes them to expand. In the time between beats, your arteries relax again. On the underside of the Apple Watch is a sensor, called a photoplethysmogram (PPG), that uses green and infrared LEDs to shine light onto your skin, and detects the small changes in the amount of light reflected back as your arteries expand and relax with each beat of your heart. Using this sensor, the Apple Watch can tell how fast your heart is beating, and how your heart rate changes over time.
But, your heart rate does not tell everything there is to know about your heart. The PPG sensor on the Apple Watch can only see what happens after each heartbeat, as blood is pumped around your body. It can’t tell you anything about what is making your heart beat, or about what happens inside your heart during each beat. An ECG is very different, and tells you a lot more!
An ECG measures the electrical activity in your heart muscles. It detects the small pulse of electricity from the sinoatrial node (the body’s natural pacemaker, which normally initiates each heartbeat) and the large electrical impulses produced as the lower chambers of the heart (the ventricles) contract and relax. By looking at an ECG, a doctor can discern a wealth of information about the health and activity of your heart muscle, much more than you can tell from your heart rate alone. ECGs are the required gold standard for diagnosis of arrhythmias and many cardiac abnormalities, and can even be used to see evidence of acute heart attacks and even events that have occurred in the past.
Research has shown that taking frequent ECGs increases the likelihood of detecting certain arrhythmias, and decreases the mean time to diagnosis.
The skeptical cardiologist has been evaluating the Kardia Band from AliveCor which allows one to record single lead medical grade ECGS on your Apple Watch. What follows is my initial experience with setting up the device and using it to make recordings.
After ordering my Kardia Band for Apple Watch on 11/30 from AliveCor the device appeared on my door step 2 days later on a Saturday giving me most of a Sunday to evaluate it.
What’s In The Box
Inside the box I found one small and one large black rubber wrist watch band
The larger one had had a small squarish silver metallic sensor and the smaller one had a space to insert a sensor. It turns out my wrist required the smaller band and it was very easy to pop out the sensor and pop it into the smaller band.
After replacing my current band with the Kardia band (requires pushing the button just below the band and sliding the old band out then sliding the new one in) I was ready to go.
The Eternal fiancée did not complain about the appearance of the band so I’m taking that to mean it passes the sufficiently stylish test. She did inquire as to different colors but it appears AliveCor only has one style and one color to choose from right now.
I have had problems with rashes developing with Apple’s rubbery band and switched to a different one but thus far the Kardia band is not causing wrist irritation.
I didn’t encounter any directions in the box or online so I clicked on the Kardia app on the watch and the following distressing message appeared.
Prior to 11/30 Kardia Band only worked in certain countries in Europe so I suspected my AliveCor app needed to be updated.
I redownloaded the Kardia app from the Apple App Store , deleted it off my Watch and reinstalled it.
I was thrilled when the app opened up and gave me the following message
However, I was a little puzzled as I was not aware that setting up Smart Rhythm was a requirement to utilize the ECG recording aspect of Kardia Band. Since I have been granted a grandfathered Premium membership by AliveCor I knew that I would have access to Smart Rhythm and went through the process of entering my name and email into the Kardia app to get this started.
Alas, when the Watch Kardia app was accessed after this I continued to get the same screen. Clicking on “need help” revealed the following message:
Bluetooth was clearly on and several attempts to restart both the watch and the iPhone app did not advance the situation.
I sent out pleas for assistance to AliveCor.
At this point the Eternal Fiancee had awoken and we went to Sardella for a delightful brunch . I had this marvelous item:
Later on that day I returned to my Kardia Band iPhone and deinstalled, reinstalled , reloaded and restarted everything.
The First Recording
At this point it worked and I was able to obtain my first recording by pushing the record ECG button and holding my thumb on the sensor for 30 seconds.
I’ve made lots of recordings since then and they are good quality and have accurately recognized that I am in normal sinus rhythm.
The Smart Rhythm component has also been working. Here is a screen shot of today’s graph.
You’l notice that the Smart Rhythm AI gave me a warning sometime in the morning (which I missed) as it felt my rhythm was abnormal. I missed making the recording but am certain that I was not in afib.
Comparison of the Kardia Band recording (on the right) versus the separate Kardia device recording (on left) shows that they are very similar in terms of the voltage or height of the p waves, QRS complexes and T waves.
I felt a palpitation earlier and was able to quickly activate the Kardia Watch app and make a recording which revealed a PVC.
In summary, after some difficulty getting the app to work I am very pleased with the ease of recording, the quality of the recording and the overall performance of Kardia Band. The difficulties I encountered might reflect an early adoption issue which may already be resolved. Please give me feedback on how the device set up worked for you.
I’ll be testing this out on patients with atrial fibrillation and report on how it works in various situations in future posts.
After more experience with the Smart Rhythm monitoring system which I think could be a fantastic breakthrough in personal health monitoring I’ll give a detailed analysis of that feature.
AliveCor has finally gotten approval from the FDA to release its Kardia Band in the United States.
The skeptical cardiologist is quite excited to get his hands (or wrist) on one and just gave AliveCor $199 to get it.
The device incorporates a mobile ECG sensor into a wrist band that works with either 42 or 38 mm Apple watches. I’ve written extensively about AliveCor’s previous mobile ECG product (here and here) which does a good job of recording a single lead ECG rhythm strip and identifying atrial fibrillation versus normal rhythm,
Hopefully, the Kardia Band will work as well as the earlier device in accurately detecting atrial fibrillation.
According to this brief video to make a recording you tap the watch screen then put your thumb on the sensor on the band.
The app can monitor your heart rate constantly and alerts you to make a recording if it thinks you have an abnormal rhythm.
I was alerted to the release of Kardia by Larry Husten’s excellent Cardio Brief blog and in his post he indicates that the alert service , termed Smart Rhythm, requires a subscription of $99 per year.:
AliveCor simultaneously announced the introduction of SmartRhythm, a program for the Apple Watch that monitors the watch’s heart rate and activity sensors and provides real-time alerts to users to capture an ECG with the Kardia Band. The program, according to an AliveCor spokesperson, “leverages sophisticated artificial intelligence to detect when a user’s heart rate and physical activity are out of sync, and prompts users to take an EKG in case it’s signaling possible abnormalities like AFib.”
The Kardia Band will sell for $199. This includes the ability to record unlimited ECGs and to email the readings to anyone. The SmartRhythm program will be part of the company’s KardiaGuard membership, which costs $99 a year. KardiaGuard stores ECG recordings in the cloud and provides monthly summary reports on ECGs and other readings taken.
AliveCor tells me my Kardia Band will be shipped in 1-2 days and I hope to be able to give my evaluation of it before Christmas.
Please note that I paid for the device myself in order to avoid any bias that could be introduced by receiving largesse from AliveCor.
N.B. Larry Husten’s article includes some perspective and warnings from two cardiologist and can be read here.
Another article on the Kardia Band release suggests that the Smart Rhythm program at $99/ year is a requirement.
Perhaps, AliveCor’s David Albert can weigh in on whether the annual subscription is a requirement for making recordings or just allows the continuous monitoring aspect.
The Skeptical Cardiologist is a strong proponent of empowering patients with atrial fibrillation by utilizing personal cardiac rhythm devices such as Afib Alert or AliveCor’s Kardia.
I’ve written about my experiences with the initial versions of the Kardia mobile ECG device and the service it provides here and here.
I have been monitoring dozens of my afib patients using AliveCor’s Physician Dashboard.
Recently AliveCor changed fundamentally the way their app works such that for new users much of the functionality I described in my previous posts now requires subscribing to their Premium service which costs $9.99 per month or $99 per year.
What Has Changed With The Kardia App
The Kardia device which works with both iOs and Android smart
phones is unchanged and still generates a “medical-grade” single lead rhythm strips which appears within the Kardia app.
The app still is reasonably accurate at identifying atrial fibrillation or normal heart rhythms and offers a fee-based service for interpretation of unclassified ECGs.
However, for new purchasers of Kardia, the capability to access, email or print prior ECG recordings has gone away. Prior to March of this year, Kardia users could access prior ECG tracings which were stored in the cloud by touching the “Journal” button on the app. These older tracings could be emailed and they were available through the cloud for a physician like myself to review at any time.
Now new Kardia purchasers will find that when they make an ECG recording they have the option to email a PDF of the ECG but once they hit the DONE button it is gone and is not stored anywhere.
For my patients purchasing after March, 2017 this means that unless they purchase Kardia Premium service I will not be able to view their ECG recordings online.
An AliveCor account executive summarized for me the changes as follows:
We added a significant number of features over the past year and a half, and grandfathered all users on March 16th, 2017. New users now have the option to download and use Kardia for free, but the premium services are $9.99/mo or $99/year. Kardia Premium allows unlimited storage and history of their EKGs, summary reports with longitudinal data, blood pressure monitoring and tracking weight and medication.
Why Journal Functionality Is Important
If you purchased your AliveCor/Kardia device prior to March 16th, 2017 ago the journal functionality still works. Let’s call such customers “Journal Grandfathered”.
This Journal functionality is important in a number of ways:
My Journal Grandfathered patients can bring their phones with them during an office visit and we can review all of their ECG tracings.
Journal gGandfathered Kardia users can email their old tracings to their physicians or to anyone they wish (even the skeptical cardiologist!). They can also print them out and save PDFs of the tracings.
I can view through my online physician account all of my Journal Grandfathered patients. This means any time a patient of mine makes a recording that is unclassified or suggests atrial fibrillation I can be notified and immediately view it online.
This fundamental change took place as AliveCor attempts to convince purchasers of the Kardia device to use their Premium service.
Why AliveCor Changed The Kardia App Function
Dr. David Albert, inventor and cardiologist and the founder of AliveCor was kind enough to talk with me about this change.
He indicates that of the 150,000 AliveCor users, 10,000 are now using the Kardia Premium service. About 20% of new users elect Kardia Premium.
Prior to the change all AliveCor users had their old ECG recordings stored in the cloud in a HIPPA compliant fashion. This free service was costing AliveCor quite a bit and the company felt it was best to switch to a subscription service to provide this secure cloud storage.
With the change to the (relatively inexpensive) subscription service, patients will get additional features. As the AliveCor account executive described:
Kardia Premium allows unlimited storage and history of their EKGs, summary reports with longitudinal data, blood pressure monitoring and tracking weight and medication.
I’ve looked at the Premium service and it seems quite useful when combined with a connected physician utilizing Kardia Pro. I’ll evaluate the Premium service and the physician Kardia Pro service further and write a full post on its features in the near future.
If you are not grandfathered and want to stick with the Basic Kardia service you still have an immensely useful and inexpensive device which allows personal detection of your cardiac rhythm. Just remember to email yourself the ECG recording you just made before you hit DONE.
The skeptical cardiologist has been testing the comparative accuracy of two hand-held mobile ECG devices in his office over the last month. I’ve written extensively about my experience with the AliveCor/Kardia (ACK) device here and here. Most recently I described my experience with the Afib Alert (AA) device here.
Over several days I had my office patients utilize both devices to record their cardiac rhythm and I compared the device diagnosis to the patient’s true cardiac rhythm.
In 14 patients both devices correctly identified normal sinus rhythm. AFA does this by displaying a green check mark , ACK by displaying the actual recording on a smartphone screen along with the word Normal.
The AFA ECG can subsequently uploaded via USB connection to a PC and reviewed in PDF format. The ACK PDF can be viewed instantaneously and saved or emailed as PDF.
Normal by AFA/Unreadable or Unclassified by AliveCor
In 5 patients in normal rhythm (NSR) , AFA correctly identified the rhythm but ACK was either unreadable (3) or unclassified (2). In the not infrequent case of a poor ACK tracing I will spend extra time adjusting the patient’s hand position on the electrodes or stabilizing the hands. With AFA this is rarely necessary.
In this 70 year old man the AFA device recording was very good and the device immediately identified the rhythm as normal.
ACK recording was good quality but its algorithm could not classify the rhythm.
A 68 year old man who had had bypass surgery and aortic valve replacement had a very good quality AFA recording with correct classification as NSR
AliveCor/Kardia recordings on the same patient despite considerable and prolonged efforts to improve the recording were poor and were classified as “unreadable”
There were 3 cases were AFA diagnosed atrial fibrillation (AF) and the rhythm was not AF. These are considered false positives and can lead to unncessary concern when the device is being used by patients at home. In 2 of these ACK was unreadable or unclassified and in one ACK also diagnosed AF.
A 90 year old woman with right bundle branch block (RBBBin NSR was classified by AFA as being in AF.
The ACK algorithm is clearly more conservative than AA. The ACK manual states:
If you have been diagnosed with a condition that affects the shape of your EKG (e.g., intraventricular conduction delay, left or right bundle branch block,Wolff-Parkinson-White Syndrome, etc.), experience a large number of premature ventricular or atrial contractions (PVC and PAC), are experiencing an arrhythmia, or took a poor quality recording it is unlikely that you will be notified that your EKG is normal.
One man’s rhythm confounded both AFA and AC. This gentleman has had atrial flutter in the past and records at home his rhythm daily using his own AliveCor device which he uses in conjunction with an iPad.
During our office visits we review the recordings he has made. He was quite bothered by the fact that he had several that were identified by Alivecor as AF but in fact were normal.
A recording he made on May 2nd at 845 pm was read as unclassified but with a heart rate of 149 BPM. The rhythm is actually atrial flutter with 2:1 block.
Sure enough, when I recorded his rhythm with ACK although NSR (with APCS) it was read as unclassified
AFA classified Lawrence’s rhythm as AF when it was in fact normal sinus with APCs.
One patient a 50 year old woman who has a chronic sinus tachycardia and typically has a heart rate in the 130s, both devices failed.
We could have anticipated that AC would make her unclassified due to a HR over 100 worse than unclassified the tracing obtained on her by AC (on the right)was terrible and unreadable until the last few seconds. On the other hand the AFA tracing was rock solid throughout and clearly shows p waves and a regular tachycardia. For unclear reasons, however the AFA device diagnosed this as AF.
Accuracy in Patients In Atrial Fibrillation
In 2/4 patients with AF, both devices correctly classified the rhythm..
In one patient AFA correctly diagnosed AF whereas ACK called it unclassified.
This patient was in afib with HR over 100. AFA correctly identified it whereas ACK called in unclassified. The AC was noisy in the beginning but towards the end one can clearly diagnose AF
In one 90 year old man AFA could not make the diagnosis (yellow)
ACK correctly identified the rhythm as AF
One patient who I had recently cardioverted from AF was the only false positive ACK. AliveCor tracing is poor quality and was called AF whereas AFA correctly identified NSR>
The sensitivity of both devices for detecting atrial fibrillation was 75%.
The specificity of AFA was 86% and that of ACK was 88%.
ACK was unreadable or unclassified 5/26 times or 19% of the time.
The sensitivity and specificity I’m reporting is less than reported in other studies but I think it represents more real world experience with these types of devices.
In a head to head comparison of AFA and ACK mobile ECG devices I found
-Recordings using AfibAlert are usually superior in quality to AliveCor tracings with a minimum of need for adjustment of hand position and instruction.
-This superiority of ease of use and quality mean almost all AfibAlert tracings are interpreted whereas 19% of AliveCor tracings are either unclassified or unreadable.
-Sensitivity is similar. Both devices are highly likely to properly detect and identify atrial fibrillation when it occurs.
-AliveCor specificity is superior to AfibAlert. This means less cases that are not AF will be classified as AF by AliveCor compared to AfibAlert. This is due to a more conservative algorithm in AliveCor which rejects wide QRS complexes, frequent extra-systoles.
Both companies are actively tweaking their algorithms and software to improve real world accuracy and improve user experience but what I report reflects what a patient at home or a physician in office can reasonably expect from these devices right now.
I’ve been evaluating the ability of a mobile hand-held ECG device called AfibAlert to detect atrial fibrillation for the last few weeks.
I found that the device made very reliable and consistent recordings of cardiac rhythm and did a reasonably good job of detecting atrial fibrillation (afib).
The device came in a plastic case with a USB cable for uploading recordings and two metal bracelets which attach to electrodes and provide an alternative recording method.
The device itself is about 6 by 3 by 1 inch.
Recordings are made by placing your thumbs on the silver/siver chloride electrodes
After a few seconds the display in the center will give heart rate and after 45 seconds the
device will make a decision about your rhythm:
If it diagnoses normal sinus rhythm a green check appears and if it diagnoses afib a red telephone appears.
If it is confused you get yellow circular arrows.
As the maker of the device explains:
Lohman Technologies’ patented algorithm analyzes the patient’s heartbeat and the appropriate icon illuminates to show what action is needed. AfibAlert’s® algorithm was validated against 51,000+ ECG strips from the MIT-BIH Atrial Fibrillation Database with known diagnosis. The Afib monitor’s results were excellent, with 94.6% accuracy in detecting the presence of arrhythmias. Each recording produces a 45-second diagnostic quality ECG rhythm strip
The device I tested does not allow you to immediately see the ECG tracing. The recordings are uploaded to a PC via USB cable and then can be viewed as a PDF document.
I made 17 recordings on patients in my office one day. The age range was 50 to 93 years and most patients were able to rapidly and easily grasp the device with thumbs appropriately positioned to make interpretable recordings.
Only 2/17 came back. yellow. In both cases, I repeated the recording and the device was able to make the correct diagnosis. Twice I got the yellow signal on an elderly, partially blind patient who had trouble keeping his thumb on the electrode.
In 15 cases of normal sinus rhythm the device correctly identified NSR.
In one case of atrial fibrillation the device correctly identified atrial fibrillation.
In one case of SR with
APCs the device
incorrectly identified afib
Overall the device correctly classified 88% of the tracings. This was superior to the device I normally utilize ( AliveCor/Kardia mobile ECG) in head to head comparison (I’ll present this comparison in a subsequent post).
My bullet points on the AfibAlert device:
-5 stars for Ease of Recording
-5 stars for Quality of recording
In all cases that uploaded, the recordings were very clear and free of artifact. The device did not upload yellow signal events and I presume more artifact is present in these recordings.
-2 stars for Convenience.
I found the software and uploading to be very awkward and slow. The company indicates new software soon to be released along with the ability to interface directly with iPads or smartphones that hopefully will improve this factor.
The inability to instantaneously view the ECG tracings means I cannot use it in my office to screen patients for arrhythmias. However, if a patient is solely using it to determine if afib is present or absent, this information is available right away.
-3 stars for Accuracy.
It does a reasonable job of identifying the patient who is in normal rhythm versus one in atrial fibrillation.
However, like AliveCor and other devices which strictly look at the variability of the pulse, it can be easily fooled by premature beats, especially when they are frequent, and inappropriately classify these as afib resulting in false positives.
In addition, when afib rates are very slow and thus much less variable it is likely AfibAlert will incorrectly classify them as normal thus resulting in false negatives.
False Negatives and False Positives
False negatives result in delayed diagnosis of afib. Patients will be falsely reassured that their rhythm is normal when it is not.
False positives result in needless anxiety and testing/treatment.
If afib monitoring devices are to be successful they have to have a very low frequency of both types of inaccuracy.
The solution to inaccuracy of interpretation, of course, is to have a cardiologist over-read the tracings.
AfibAlert recordings are available online for review by your personal physician after being uploaded. This requires your physician to have an account with AfibAlert. There is no capability for having the recordings over read by an online cardiologist for a charge.
As far as I can tell the device is only available for purchase in the US and only on the AfibAlert website.
Interestingly, you cannot purchase AfibAlert without a prescription from a physician.
Why this is mandated for AfibAlert and not AliveCor is a mystery to me.
The skeptical cardiologist received an email from the folks at AliveCor a few days ago with the subject line:
Dad’s heart matters – Kardia Mobile for Dad will give you peace of mind and make Dad happy
The email contains this image of an older well-dressed man (with lots of bling) standing in a beautiful meadow near the ocean. The man has decided to turn his back on the ocean and check his heart rhythm using the AliveCor/Kardia (AliveCor has changed the name of its ECG devices to Kardia) mobile ECG. This man is a happy dad! (Unless his heart rhythm is interpreted as atrial fibrillation. Then the beach walk is ruined.)
The email asks the question “What if Dad’s heart really was an open book?”
Uhh, he’d be dead? Clearly books don’t function well at pumping 5 or 6 liters of blood through the cardiovascular system every minute whether they are open or closed. Perhaps the question is using either the heart or an open book as a metaphor?
The advertisement goes on to suggest that I get my dad an AliveCor device for father’s day “So you always know what his heart is thinking.”
I believe this is the marketing person’s attempt to extend the metaphor of the open book, i.e., you know exactly what dad’s brain is thinking, now you can extend this knowledge to his heart. The metaphor of the heart “thinking” is quite poor but poor metaphors are the norm today.
Bad metaphors and bad writing abound on father’s day because 90 million greeting cards are purchased and given as (according to the Greeting Card Association) “a meaningful expression of personal affection for another person.” Despite the increasing use of Facebook and its ilk to transmit emotions, the Greeting Card Association assures us that “The tradition of giving greeting cards … is still being deeply ingrained in today’s youth, and this tradition will likely continue as they become adults and become responsible for managing their own important relationships.
Mobile Ecg Monitor As A Father’s Day Gift
I have to say that despite the horror of the writing in this email advertisement it got me thinking about getting my father a Kardia device. I’ve suggested previously that an AliveCor device would make a good gift for Christmas for a loved one who has intermittent unexplained palpitations or atrial fibrillation but had not considered this for my dad.
For one thing he does not possess a smart phone which is required to make the Kardia device functional. For another, he doesn’t have atrial fibrillation (that we know of. Perhaps if I knew what his heart was thinking we would find out that it likes to fibrillate late at night,)
Perhaps it’s time to upgrade my Dad to an iPhone I began thinking.
But wait! He has an iPad mini (that he seems to only use for FaceTime conversations.)
Further research reveals that Kardia is not only compatible with iPhone and Android smartphones but apparently iPads and IPod Touch.
Taking Care of Dad’s Heart
What about the rest of the slick advertising copy in my email?
And now you can know the way to help take care of it. Kardia gives Dad a medical-grade EKG in only 30 seconds. It even gives him expert analysis and tracking, with reports getting shared directly with his physician
This part is pretty clear and correct. I use Kardia daily in my office to record patient’s heart rhythm and I have a dozen patients now who make recordings outside of the office. They can have their recordings read by a random cardiologist for a fee or establish a link with me as their provider and I can review them through my account for free.
Is It The First Father’s Day Gift That Leads To More Father’s Days?
The ad ends with the remarkably brazen statement that “It’s the first Father’s Day gift that leads to more Father’s Days.”
While I find the device more helpful in many instances than current expensive and intrusive long term monitoring devices for detecting and monitoring atrial fibrillation and other abnormal heart rhythms, it is a huge leap to suggest that this translates somehow into a longer life span.
To AliveCor’s credit, despite such ridiculous marketing drivel , studies presented at the recent Heart Rhythm Society Scientific Meetings suggest:
Kardia Mobile Superior to Conventional Monitoring: Researchers at the Leeds General Infirmary found that the AliveCor monitor is superior to conventional Holter monitoring in patients with palpitations, providing a higher diagnostic yield, more detected arrhythmias, with a similar workload.
Kardia Mobile Leads to Improved Patient Compliance:Researchers at the University of Buffalo found that AliveCor provides a diagnostic yield comparable to a 30-day ambulatory looping event monitor and that the smartphone-based ECG monitor can be used as a first approach for the diagnosis of palpitations.
Kardia Mobile provided more information resulting in changes in arrhythmia patient management than traditional external event recorders in a study from researchers at the University of Miami.
AliveCor’s AF algorithm was reported to be superior by researchers at Arizona State University to the patient’s own ability to detect AF via symptoms.
But even if these studies make it to publication they don’t suggest the device provides any improved longevity. In fact, such data, do not exist for any monitoring device.
Happy Father’s Day, Dad! Don’t be surprised when we FaceTime later today that I’ve found another use for your iPad.
N.B. Clearly I receive no consulting, speaking or P.R. writing fees of any kind from AliveCor. Nor do they provide me with any free devices. What’s more, when I lose one of their devices they don’t replace it. I am totally free of any conflict of interest.